Multiple portable socket-outlets mechanical strength test device.
Mechanical Strength Test Apparatus for IEC60884 Mechanical Strength Test on Multiple Portable Socket-outlets.
Product details:Model ZLT-MS
To determine the mechanical strength test on multiple portable socket-outlets, in according to IEC60884-1 figure 29,IS1293 figure 24.
The free end of the flexible cable is fixed to a wall at a height of 750 mm above the floor, the specimen is held so that the flexible cable is horizontal and then it is allowed to fall onto a concrete floor, eight times, the flexible cable being rotated through 45°at its fixing after each fall.
After the test, the specimens shall show no damage within the meaning of this standard; in particular, no part shall have become detached or loosened.